LCD Readout Display for Vertical/ Horizontal/Frequency Measurement
TV(TV-V, TV-H) Trigger Modes
Z-Axis Input and External Trigger Input
The Economic Oscilloscope GOS-630FC establishes a brand new benchmark for similar categories. New and innovative functions, including LCD Readout Display, Frequency Counter, and Auto Time-Base setting, are provided as standard features. With a 30MHz bandwidth and valuable features, GOS-630FC is priced at the basic Oscilloscope level. 1mV/div High Vertical Sensitivity is able to capture small signal variations in the input source. Additional functions, such as XY mode display, MAG function (Time based Magnified), and Z-axis Input, extend the application range of GOS-630FC to both education and industry fields.
The added values without being converted into extra cost make the GOS-630FC the most beneficial choice among the entry-level analog oscilloscopes available in today’s market.
Digital / Analog
Number of Channel
1mV ~ 5V
Time Base Range
0.2μs ~ 0.5s
Input Impedance Selection
Panel Setup Memory
V, Only Provided Time-Base Function.
Battery Power Operation
AC 115/230V ±15% Selectable
More product information please see the download page.
User manual x1 Power cord x1 GTP-060A-4: 60MHz (10:1/1:1) Switchable passive probe (one per channel)
Optional Accessories GTL-110 Test lead, BNC-BNC heads GTC-001 Instrument cart, 450(W) x 430(D)mm (120V input socket) GTC-002 Instrument cart, 330(W) x 430(D)mm (120V input socket)
30MHz, 2-Channel, Analog Oscilloscope
Manuals & Guides
GOS-630FC User Manual (pdf)
Brochure & Datasheet
In a workshop environment with mostly artificial lighting, the cursor display text is reasonably legible. However ... in a room with more ambient light, the cursor readout can be too faint to read (and please note that there is no direct sunlight on the unit shown below, and the text display brightness is at maximum) The questions... 1) Is there any way to boost the brightness of the cursor information display? 2) If not, does the 6051 (or any other) model have a brighter text readout?
Most of analog (CRT) oscilloscope under the strong light environment usually its result of shown not very good, certainly this includes our GOS-6031 too.
In order to solve the issue, we offer the following solution as below;
1.Adjust the trace intensity in the front panel 2. Adjust the readout intensity in the front panel
The pushbutton is for selecting the text readout intensity function or scale illumination function, and indicates the letter “TEXT” or “ILLUM” in the readout. Press the pushbutton for the following sequences:
“TEXT” --> “ILLUM” --> “TEXT” (Step 1)
The TEXT/ILLUM function are associated the VARIABLE(Step 2) control knob. Turning the knob clockwise to increase the text intensity or scale illumination, while turning the knob counterclockwise to decrease it.
If the methods of the above are all unable to reach the anticipated, we think perhaps 50MHz model will be a good choice too, because the luminance of CRT is sure to batter than 30MHz
For analog oscilloscope, some got CH1 signal output, and some have Trigger signal output, are both same meaning?
From following picture you can find there are totally different signal. CH1 signal output will be same with CH input signal, and Trigger signal output will be like pulse signal.
Why is the low frequency of the oscilloscope CRT brightness so dark? it can not be the appropriate observed signal.
Because CRT of 30MHz does not have Third Accelerating Anode (P3) to accelerate the high voltage, so the luminance of trace should not be too bright.
The scope display changes with instrument attitude. With inputs grounded the trace positions change in X and Y axis and in rotation by approx. 2mm as the instrument is moved from face vertical to face horizontal position.
Because of CRT relation of physical characteristics so the trace of the oscilloscope is very easy to be influenced, so when you change the direction of the oscilloscope, usually the trace will be influenced by magnetic field of the earth, which is the reason why the oscilloscope usually all have a Trace Rotation function.
I need to know if the GOS-630FC is a real analog scope with DSO capabilities. My primary concern is that the scope must be able to display complex waveforms accurately and properly (RF eye-pattern of a CD/DVD player.
GOS-630FC is an analog type oscilloscope with digital control, but not a digital type oscilloscope (DSO). For CD/DVD signal measure application purpose, we think using analog type oscilloscope is better than using digital type oscilloscope. Analog type oscilloscope has faster update rate and real time waveform display, which digital oscilloscope (DSO) can not perform as well.
However, if you must get eye-pattern, the best way would be using a very high-end DSO (much more accurate). But unfortunately, neither GOS-630FC nor GDS-2000A (GWInstek DSO) can do eye-pattern.